TSP-02 table for drying (heating) TLC plates
The heating of TLC plate before application of samples allow to minimize the edge- effects, obtain small size of drops and bars.
Digital control and indication, accuracy of keeping constant temperature is +\- 1 deg Celsius. Splash- protected control keys.
|Maximal size of TLC plate||up to 220 x 220 mm.|
|Temperature range set-up – from ambient temperature||to 60 C Celsius|
|Time of heating from 20 to 50 C||not over 3 min|
|Indication of “Heating” and “Pre-set T keep”||yes|
|Overall size||230x280x15 mm|
|Power supply – wall plug – in adaptor||19 V \3 A|
|Consumed power||not over 40 W|
Keeping the sorbent layer at higher temperatures leads to better separation of fractions, and increase the repeatability and resolution of TLC. The TCP-02 plate is also used for pre- activation of sorbent, and drying the samples after the process.